- Open Access
Development and application of a spectrophotometric method in quality evaluation of benzimidazole anthelminthics in Nairobi city county
Future Journal of Pharmaceutical Sciences volume 6, Article number: 74 (2020)
Neglected tropical diseases (NTDs) are a group of communicable diseases which are prevalent in the tropics affecting more than one billion people. Treatment and prevention of these infections is very costly to developing economies. Helminthiases are classified among NTDs. The communities afflicted are poor and have limited access to essential resources for their livelihood. Poor-quality drugs for NTDs may lead to death or prolonged treatment without achieving the desired results. The limited resources used in purchasing poor-quality drugs will therefore be wasted instead of being put to good use.
Most of the methods available for the analysis of benzimidazole anthelminthics utilize high-performance liquid chromatography. They are therefore time consuming, require sophisticated and expensive equipment, utilize rare and expensive reagents and solvents, and call for skilled personnel. A simple, rapid, and inexpensive ultraviolet spectrophotometric method of analysis would therefore come in handy especially in the analysis of many samples as occurs during post-authorization market surveillance for quality.
The suitable solvent for the spectroscopic analysis was established as 0.1 M methanolic HCl. The wavelength of analysis was set at 294 nm. Upon validation, the method was found to have good linearity. The range over which linearity was established was way beyond the 80 to 120% of the working concentration specified by the ICH. The method exhibited good precision.
Out of 32 commercial samples analyzed, five (15.6%) did not comply with compendial specifications. Intra-brand batch variation was also observed. Out of three batches of product A002T analyzed, one did not comply with compendial specifications.
A major limitation in the analysis of benzimidazole anthelminthics is the lack of reliable, simple, rapid, and low-cost methods of analysis with high throughput. The developed method serves to fill this gap. It can be used in the analysis of raw materials and finished products. It can also be used in the establishment of the quality of products prior to registration. The method will prove very useful in post-market surveillance of quality of benzimidazole anthelminthics.
Benzimidazole anthelminthics are broad-spectrum anthelminthics widely used both in human and veterinary medicine. Helminth infections are classified as neglected tropical diseases (NTDs) by the World Health Organization (WHO) . Most of the drugs that are available for the treatment of helminthic infections in humans were first developed as veterinary medicines . Benzimidazole anthelminthics, mainly albendazole and mebendazole (Fig. 1), play a key role in the treatment of soil-transmitted helminthiases (including ascariasis, trichuriasis, hookworms, threadworm, and pinworm infections) . Further, albendazole is the first-line treatment for hydatid disease . Albendazole is particularly preferred because of its convenience of administration as a single dose in most infections. Affordable generic formulations are also available for both albendazole and mebendazole.
Due to a dearth of effective anthelminthics for human use, the quality of the few available anthelminthics should be guarded for therapeutic success. Various analytical methods have been reported for the analysis of albendazole and mebendazole in bulk and dosage forms. High-performance liquid chromatographic (HPLC) methods, including those in the United States pharmacopeia (USP) and British Pharmacopoeia (BP), have been widely developed for these drugs [5, 6]. Very little effort has been directed towards the development of an ultraviolet (UV) spectroscopic method of analysis despite its many advantages. Compared to HPLC, UV spectroscopy is faster and requires less analyst skill and the equipment is less expensive and easier to operate and maintain. Additionally, with portable UV spectrophotometers, analysis can be performed in areas remote from the major laboratory when necessary. To date, only one UV spectroscopic method developed by Agrawal et al. has been reported .
The biochemical target for benzimidazole anthelminthics is the β-tubulin, a cytoskeletal protein which is a building block of microtubules present in all eukaryotic cells. Microtubules are critical cytoskeletal polymers which are made of repeating α- and β-tubulin dimers. Microtubules are involved in cellular morphology, cell transport, cell motility, and cell division 
Methanol of HPLC grade (Finar Ltd, India) was obtained from Chemoquip Ltd, Nairobi. Analytical grade concentrated hydrochloric acid (HCl), sodium lauryl sulfate (SLS), and albendazole and mebendazole working standards were provided by the Drug Analysis and Research Unit (DARU) of the Department of Pharmaceutical Chemistry, University of Nairobi. Commercial pharmaceutical products containing albendazole or mebendazole active pharmaceutical ingredient (API) were acquired from wholesalers in the Central Business District (CBD) and the outskirts of the city of Nairobi, Kenya. Throughout the period of the study, nine albendazole and two mebendazole brands were analyzed.
All weights were taken using a Sartorius top-loading electronic weighing balance (Sartorius GMBH, Germany). Absorbance readings were read on a Genesys 10S UV-Vis Spectrophotometer (ThermoFisher Scientific, China).
A Merck Hitachi HPLC machine (Hitachi Ltd, Tokyo, Japan), with a Varian HPLC column, 250 × 4.0 mm, 5 μm LiChristopher 100-5 RP 18 end capped, kindly availed by the National Quality Control Laboratory (NQCL), Nairobi, Kenya, was used for the orthogonal analysis of commercial samples. It was equipped with an L-7100 low pressure quaternary pump, an L-m7200 autosampler, an L-7400 variable UV detector set at 308 nm, an L-7350 thermostatic column oven maintained at 40 °C, and an L-7000 computer interphase.
A method from literature developed by Agrawal et al.  was adapted for the analysis of both albendazole and mebendazole at a single wavelength using a common solvent. The key considerations in development of the method included choice of solvent in which both albendazole and mebendazole exhibited adequate solubility, determination of optimal wavelength of analysis, and determination of suitable working concentration. To make the analytical process simple, analysis needed to be performed at a common wavelength at which both APIs showed adequate absorbance with minimal interference from excipients, related substances, and degradation products possibly present in analytical samples. A working concentration within the linear range of absorption signal of both APIs also had to be determined.
Determination of a suitable solvent
Two solvents were investigated for the dissolution of both the bulk APIs and the commercial samples. These were 0.1 M HCl containing 0.05% SLS and 0.1 M methanolic HCl. The former had been used in the Agrawal et al. method  while the latter was used by Al-Kurdi et al. .
Choice of wavelength of analysis
To decide on a single wavelength for both analytes, the UV spectra of each API at a nominal concentration of 12 μg/mL in 0.1 M methanolic HCl were run independently between 200 and 400 nm. The two spectra were then overlaid. Two wavelengths (233 and 294 nm) were initially chosen for further investigation.
Choice of working concentration
The appropriateness of a concentration of 12 μg/mL as used by Agrawal et al.  was investigated and was found to fall within the linear range for both APIs.
After the preliminary investigations (“Determination of a suitable solvent,” “Choice of wavelength of analysis,” and “Choice of working concentration” sections), optimal conditions for the method were suggested as UV absorbance of a 12-μg/mL solution of each API in 0.1 M methanolic HCl and measured at 294-nm wavelength. This method was taken through a validation process to assess its suitability.
Linearity and range
A 1.0-mg/mL stock solution of each of the two APIs was prepared by weighing 50 mg of the respective API into a 50-mL volumetric flask, dissolving in minimum 0.1 M methanolic HCl, and the solution made to volume with the same solvent. Working solutions were prepared by transferring aliquots of the stock solution into 25-mL volumetric flasks and making to volume using 0.1 M methanolic HCl producing ten solutions of 4, 8, 12, 16, 20, 24, 28, 32, 36, and 40 μg/mL nominal concentrations. This represented a range of 33.3 to 333.3% of the working concentration. The absorbances of these solutions were read at 294 nm, and the data obtained plotted using a Microsoft Excel spreadsheet and subjected to linear regression analysis.
Repeatability and intermediate precision were determined in this study as outlined in the sections “Repeatability” and “Intermediate precision.” Reproducibility was not determined as the study did not involve a collaborating laboratory.
About 50 mg of each API was weighed into a 50-mL volumetric flask, dissolved in minimum 0.1 M methanolic HCl, and made to volume with the same solvent. A 0.3-mL aliquot of this solution was transferred to a 25-mL volumetric flask and made to volume with the same solvent to give a final solution. Absorbance of the test solution was determined on the same day six times at 294 nm. The standard deviation, relative standard deviation, and coefficient of variation (COV) of these data were then calculated.
The procedure for the determination of repeatability (“Repeatability” section) was followed after several days.
The accuracy of the method was established by adding a known amount of the analyte (API) to a solution of a commercial product whose API concentration was 80, 100, and 120% of the working concentration (12 μg/mL) . The percentage recovery of the analyte in each solution was then determined. The determinations were done in triplicate.
Orthogonal HPLC analysis
To compare the reliability and accuracy of the developed method with that of a validated method in routine use, the HPLC procedure for the analysis of albendazole as described in the USP 2018 was used. The suspension dosage form of one of the commercial products was analyzed. The results obtained from both methods were then compared.
The process of testing for accuracy (“Accuracy” section) involving the analysis of the API in the presence of excipients, possible related compounds, and degradation products was additionally used to assess the specificity of the developed method.
As a measure of sensitivity, the limits of detection and quantitation (LOD and LOQ) were determined by computing the standard deviation (σ) of the response and the slope (S) of the linearity plot . The standard deviation was determined by measuring the absorbance of the blank (0.1 M methanolic HCl) six times and calculating the standard deviation of the responses.
Analysis of commercial samples
The linear plots used in the determination of linearity and range (“Linearity and range” section) were also used as the calibration curves for content determination of both APIs.
Tablet dosage forms
Twenty tablets were accurately weighed and pulverized to a fine powder. An amount of the powder equivalent to 50 mg of the respective API was accurately weighed into a 50-mL volumetric flask. About 25 mL of 0.1 M methanolic HCl was added and the mixture shaken to dissolve. The solution was ultrasonicated for 5 min and made to volume with the same solvent and the solution filtered. A 0.3-mL aliquot of the filtrate was pipetted into a 25-mL volumetric flask and made to volume with the same solvent. The absorbance of this solution was read at 294 nm. The samples were prepared in triplicate.
Suspension dosage forms
An amount of the suspension equivalent to 50 mg of the respective API (as determined by the use of a density bottle) was accurately weighed into a 50-mL volumetric flask. A minimum amount of 0.1 M methanolic HCl was added and the flask shaken to dissolve. The solution was ultrasonicated for 5 min and made to volume with 0.1 M methanolic HCl and the solution filtered. A 0.3-mL aliquot of the filtrate was pipetted into a 25-mL volumetric flask and made to volume with the same solvent. The absorbance of this solution was read at 294 nm. The samples were prepared in triplicate.
Choice of solvent
Both APIs were found to have better solubility in 0.1 M methanolic HCl than in 0.1 M HCl containing 0.05% SLS. Therefore, 0.1 M methanolic HCl was used for further development of the analytical method.
Choice of wavelength of analysis
The optimal wavelength for API signal detection was determined to be 294 nm. At this wavelength, least interference was exhibited by other possible substances present in commercial products as compared to 233 nm.
Linearity and range
The data obtained (Figs. 2 and 3) showed good linearity (between 33.3 and 333.3% of the working concentration) for each of the two APIs. The coefficient of determination, R2, was 0.9989 for both APIs.
The CV of albendazole was 0.184% and 0.0% for mebendazole (Table 1). Since the CVs were below 2% , the developed method exhibited good repeatability.
The data for recovery studies is presented in Table 2.
The average recovery for the three levels for albendazole was 102.3% and 104.2% for mebendazole.
The Food and Drug Administration (FDA) of the USA requires that the recovery should be 100 ± 2% at each concentration over the range of 80 to 120% of the working concentration .
Though the results (102.3% and 104.2% recovery for albendazole and mebendazole, respectively) were slightly above the upper limit at some concentrations for both APIs, the developed method exhibited acceptable accuracy. The method is more accurate for albendazole than mebendazole.
Orthogonal HPLC analysis
The percentage label claim for the selected analyzed product using HPLC was determined to be 106.6%. This compared well with the 107.3% percentage content obtained with the developed method. This further confirms the accuracy of the developed method.
For the developed method, the results of the recovery studies indicate that the method is capable of discriminating the analyte in the presence of the components likely to be present in the commercial products including excipients, related substances, and products of degradation. The method was therefore found to be specific for albendazole and mebendazole.
The LOD and LOQ results were albendazole, LOD = 43.5 ng/mL and LOQ = 131.9 ng/mL; for mebendazole, LOD = 30.6 μg/mL and LOQ = 92.7 ng/mL. It is worth noting that while the working concentration is stated in micrograms, the LOD and LOQ are stated in nanograms. These relatively low figures imply high sensitivity of the developed method for both APIs .
Assay of commercial products
Out of the 32 samples analyzed, five samples (15.6%) did not comply with compendial specifications. From the information gathered in the field, albendazole is the more popular anthelminthic compared to mebendazole. This is because it is administered as a single dose and several low-cost generic brands are available. It is therefore of great concern when a low-cost generic brand fails to conform to compendial specification since these drugs are more affordable and therefore mostly used by a greater percentage of the population. It came as a surprise that a suspension of the innovator product of mebendazole had an overage of the API hence did not conform to compendial specification. This is because the innovator product is usually used as the gold standard when studying the pharmaceutical equivalence of generic products [12, 13]. Also, inter-batch variation was observed with product A002T, a tablet dosage form of albendazole which is a popular anthelminthic, with one batch of the product having an API overage thus not complying with the compendia specifications.
Post-marketing surveillance is an essential component of drug discovery and development and should routinely be performed by the pharmaceutical companies, drug regulatory authorities, and analytical laboratories to ascertain that marketed medicines meet compendial requirements and hence assure their continued usefulness [14, 15]. In low-resourced jurisdictions such as Africa, there is a dearth of more sophisticated analytical techniques and skilled manpower. Cheaper but accurate methods of analysis would therefore be more appropriate. Bearing this in mind, a UV spectroscopic method for the analysis of benzimidazole anthelminthics albendazole and mebendazole was developed and validated. When the method was applied to commercial products, it was found to work as reliably as it did for the bulk raw material. The method was found to be comparable to other validated compendial methods. It could therefore find applicability in the analysis of bulk raw materials and finished products in manufacturing establishments. Because of its simplicity, it can be particularly useful for post-marketing surveillance of quality by regulatory bodies. With portable UV spectrophotometers being available, kits can easily be developed for field analysis of samples on site.
Availability of data and materials
The data sets used and generated during the current study are available from the corresponding author on reasonable request.
Active pharmaceutical ingredient
Central business district
Coefficient of variation
Drug analysis and research unit
Food and drug administration
High-performance liquid chromatography
Limit of detection
Limit of quantitation
National quality control laboratory
Neglected tropical disease
Relative standard deviation
Sodium lauryl sulfate
World health organization
Hotez PJ, Brindley PJ, Bethony JM et al (2008) Helminth infections: the great neglected tropical diseases. J Clin Invest 118:1311–1321
Holden-Dye L, Walker RJ (2007) Anthelmintic drugs. WormBook 2:1–13
Bethony J, Brooker S, Albonico M et al (2006) Soil-transmitted helminth infections: ascariasis, trichuriasis, and hookworm. Lancet 367:1521–1532
Cook GC (1990) Use of benzimidazole chemotherapy in human helminthiases: indications and efficacy. Parasitol Today 6:133–136
Al-Kurdi Z, Al-Jallad T, Badwan A, Jaber AMY (1999) High performance liquid chromatography method for determination of methyl-5-benzoyl-2-benzimidazole carbamate (mebendazole) and its main degradation product in pharmaceutical dosage forms. Talanta 50:1089–1097
De Ruyck H, Daeseleire E, De Ridder H, Van Renterghem R (2002) Development and validation of a liquid chromatographic–electrospray tandem mass spectrometric multiresidue method for anthelmintics in milk. J Chromatogr A 976:181–194
Agrawal VK, Chaturvedi S, Gupta A (2015) Simple and precise UV spectrophotometric method development for estimation of albendazole for dissolution study. Int J Pharm Sci Drug Res. 2015;7(1):120–2.
Nogales E (2001) Structural insights into microtubule function. Annual review of biophysics and biomolecular structure 30:397–420
Guideline IHT (2005) Validation of analytical procedures: text and methodology Q2 (R1). International conference on harmonization, Geneva, Switzerland
Shabir GA (2005) Step-by-step analytical methods validation and protocol in the quality system compliance industry. J Validation Technol 10:314–325
Shabir GA (2003) Validation of high-performance liquid chromatography methods for pharmaceutical analysis: understanding the differences and similarities between validation requirements of the US Food and Drug Administration, the US Pharmacopeia and the International Conference on Harmonization. Journal of chromatography A 987:57–66
Zuluaga AF, Agudelo M, Rodriguez CA, Vesga O (2009) Application of microbiological assay to determine pharmaceutical equivalence of generic intravenous antibiotics. BMC Clin Pharmacol 9:1
Vesga O, Agudelo M, Salazar BE et al (2010) Generic vancomycin products fail in vivo despite being pharmaceutical equivalents of the innovator. Antimicrob Agents Chemother 54:3271–3279
Alfadl AA, Abdoon S, Elamin M, Elnabi NG (2006) Quality of antimalarial drugs in Sudan: results of post-marketing surveillance. Sudanese J Public Health 1:108–111
Alomi YA, Kamal E, Alomi K (2016) National Drug Quality Reporting System at Ministry of Health in Saudi Arabia. J Pharmacovigilance 4:2
The authors wish to acknowledge the assistance accorded by Dr. H. K. Chepkwony and Dr. Mwaura of the National Quality Control Laboratory who facilitated the performance of orthogonal HPLC analysis using the facilities at their laboratory.
This study was funded by funds kindly availed by the World Bank Through MAPRONANO ACE, Makerere University, Kampala, Uganda. The funding authority played no role whatsoever in the conduction of the study, data analysis, or writing of the manuscript.
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Murage, J.K., Amugune, B.K., Njogu, P. et al. Development and application of a spectrophotometric method in quality evaluation of benzimidazole anthelminthics in Nairobi city county. Futur J Pharm Sci 6, 74 (2020). https://doi.org/10.1186/s43094-020-00103-4
- Benzimidazole anthelminthics
- UV spectroscopy