- Open Access
Analytical method development and validation for simultaneous estimation of monoammonium glycyrrhizinate and sennoside-B in polyherbal laxative tablet using RP-HPLC
Future Journal of Pharmaceutical Sciences volume 7, Article number: 49 (2021)
Standardization of polyherbal medicine though being the need of the hour is a toilsome task. Among the various methods employed for quality control and standardization of polyherbal medicine, phytochemical profiling is of utmost importance as it signifies the quality as well as efficacy of the medicine. The work was aimed to develop and validate a simple, quick, and accurate RP-HPLC technique for simultaneous assessment of monoammonium glycyrrhizinate and sennoside-B in a polyherbal laxative tablet. The phytomarkers were effectively quantified by RP-HPLC system on C18 analytical column using gradient mobile phase consisting of phosphate buffer to acetonitrile with the detector wavelength set at 254 nm. This developed method was validated by determination of parameters like accuracy, linearity, precision, limit of detection, and quantification as well as robustness according to ICH guidelines.
Calibration curve of both phytomarkers showed excellent linear correlation coefficients. LOD and LOQ were also calculated by equation. Precision studies were carried out using intra-day and inter-day intervals and RSD values were found to be less than 2.00%. The method was found to be accurate, which was evident from 98.96 to 101.39% and 99.17 to 100.67% recovery of monoammonium glycyrrhizinate and sennoside-B, respectively, when the formulation was spiked with the respective phytomarkers.
The validated method can be employed as standardization tool for herbal formulations with accuracy and precision. The developed method will assist in maintaining the good quality and batch to batch uniformity of polyherbal formulations containing Yashtimadhu, Swarnpatri, and Aragvadha as active ingredients.
Herbs and products containing herb(s) are marketed for several purposes. It is evident that moreover 80% of the entire world population relies on herbal medicines and products made from for healthy living . Plant materials and preparations are manufactured and sold most commonly in the pharmaceutical market for treating diseases and promoting public health due to its virtues like minimum cost, potency and efficiency, improved tolerance, effectivity in chronic diseases, widespread availability, more protection, fewer side-effects, complete accessibility, and recyclable. Lack of quality in medicines leads to disappointment of consumers and also can lead to fatal consequences.
Standardization or quality control of herbal formulations is significant to determine its quality standards on basis of the concentration of actives along with phytochemical, chemical, physical, and in vivo and in vitro parameters. The acceptability in modern system of medicine can be justified by the quality assessment of herbal formulations as the scope for variation in different batches of herbal medicine is obvious [2,3,4]. To ensure that the consumers obtain the medication which give assurance of safety, purity and potency along with good efficacy are the primary responsibility of producer and such processes rationally would apply to all the medication types of allopathy or the traditional systems .
A commercial ayurvedic laxative formulation, Vasulax, was selected for development of simultaneous estimation of phytoconstituents present in it by reverse phase high-performance liquid chromatography (RP-HPLC). Vasulax is mainly composed of Yashtimadhu, Swarnpatri, Aragvadha, and Sunthi along with excipients. Glycyrrhiza glabra L. (Liquorice) is known as yashtimadhu, one of the widely known herbs from ancient time of Ayurveda as a medicine. Traditionally, yashtimadhu is reported as expectorant, demulcent, sweetener, anti-tussive, and laxative . The chief sweet tasting and active constituent in the yashtimadhu is monoammonium glycyrrhizinate. Swarnpatri commonly known as Cassia angustifolia M. Vahl (Senna) is an important plant drug in traditional and modern medicine for the management of constipation [7,8,9]. Sennosides are the major anthraquinone glycosides accountable for purgative property of Swarnpatri. A range of poly-herbal formulations comprising swarnapatri leaves are existing in India for relief of constipation and allied troubles. Aragvadha, known as Cassia fistula L. (Amaltas), belongs to family Leguminosae reported to possess laxative effect [10, 11]. This plant is also reported to contain sennoside glycoside along with other chemical constituents.
For quantification purposes, monoammonium glycyrrhizinate was selected as a phytomarker of Yashtimadhu and sennoside-B was selected as a phytomarker of Swarnpatri and Aragvadha.
There are reported methods available in literature for determining sennoside-B and monoammonium glycyrrhizinate independently and in blend with other drugs from swarnapatri, yashtimadhu, and aragvadha [12,13,14,15,16,17]. On the other hand, there is no RP-HPLC technique documented for the simultaneous estimation of mentioned combined phytomarkers in any dosage forms. Accordingly, the present work was aimed to develop a validated RP-HPLC method for simultaneous determination of sennoside-B and monoammonium glycyrrhizinate in Vasulax tablet.
Acetonitrile—gradient grade for liquid chromatography (Merck, India)
Water—chromatography grade (Merck, India)
Potassium dihydrogen orthophosphate—analytical grade (Merck, India)
Ortho-phosphoric acid—chromatography grade (Merck, India)
Monoammonium glycyrrhizinate—purity > 99%, analytical standard (Natural Remedies, India)
Sennoside-B—purity > 99%, analytical standard (Natural Remedies, India)
HPLC—prominence UFLC (Shimadzu, Japan)
Pump—quaternary gradient LC 20 AD (Shimadzu, Japan)
Injector—auto sampler SIL 20 AC (Shimadzu, Japan)
Column oven—CTO 10 AS (Shimadzu, Japan)
Column—C18, 100 A°, 5 μm, 4.6 × 250 mm (Shimadzu, Japan)
Detector—PDA detector SPD M 20 A (Shimadzu, Japan)
Preparation of phosphate buffer
Accurately weighed 0.07 g of anhydrous potassium dihydrogen orthophosphate was transferred to 500 mL volumetric flask and approx. 450 mL chromatography grade water was added into it. This mixture was kept for ultra-sonication for 10 min for complete dissolution. 0.25 mL of orthophosphoric acid was added to it and finally volume was made up to 500 mL using water. The resulting solution was filtered through 0.22-μm nylon filter paper using vacuum filtration unit.
Preparation of reference standards solution
The solution of standard was made by dissolving accurately weighed 2.5 mg of monoammonium glycyrrhizinate and 2.5 mg of sennoside-B in a 5-mL volumetric flask and make up the volume to 5 mL using chromatography grade water. The resulting solution was sonicated for 5 min to aid dissolution and then filtered through 0.22-μm nylon syringe filter assembly.
Preparation of sample solution
Accurately weighted and powdered 2 g tablet formulation was taken to 100 mL volumetric flask and volume was adjusted to 100 mL using water of chromatography grade. The resulting solution was sonicated for 5 min and then filtered through 0.22-μm nylon syringe filter assembly.
Analytical method development
Various steps involved in the method development were optimized for accurate and precise separation of selected phytomarkers. Several parameters were studied to select optimal conditions for selection of extracting solvent, selection of column phase, selection of mobile phase, and selection of detecting wavelength. After taking several trials, the optimized chromatography method which shows optimum peak resolution and good symmetry for the simultaneous estimation of both phytomarkers is shown in Table 1.
Validation of developed method
Validation of the proposed method was performed as per ICH guideline  which includes specificity, precision, linearity, accuracy, robustness, limit of detection (LOD), limit of quantitation (LOQ), and system suitability.
Specificity is the capability to review unequivocally the analyte in the existence of components that may be anticipated to be present. To evaluate specificity of the method, two samples were prepared of which one contain placebo formulation (unspiked) and in another sample; placebo were spiked with 2 phytomarkers. The specificity was evaluated by comparing the chromatograms obtained from unspiked samples with those from samples spiked with standards individually.
Analytical procedure’s ability to attain test results that are directly corresponding to the concentration of analyte present in test sample is known as linearity. One hundred micrograms per milliliter stock solution of both phytomarkers was prepared by dissolving accurately weighted 10 mg monoammonium glycyrrhizinate and 10 mg sennoside-B in 100 mL volumetric flask. Dilution was performed to achieve final concentration of 5–15 μg/mL for both the phytomarkers. The linearity was obtained from calibration curves made by plotting average peak area against concentrations of the standards.
Detection limit and quantitation limit
The limits of detection (LOD) and limits of quantitation (LOQ) were used to evaluate the sensitivity of the developed method. For the determination of LOD and LOQ, linearity of the standards was performed three times to obtained standard deviation of the intercept (SD) and slope of the regression equation (S) value. LOD and LOQ were determined by the standard deviation method and calculated as follows:
Limits of detection = 3.3 × SD/S and Limits of quantitation = 10 × SD/S
The nearness of conformity amid a series of measurements attained from multiple sampling of the similar homogeneous sample under the stipulated circumstances is called precision of the analytical method. Precision was evaluated in terms of intra-day precision and inter-day precision (ruggedness). The intra-day precision was performed by determining response of sample 6 times in a day. The inter-day precision analyses were performed by determining the subsequent responses on 6 dissimilar days. The samples were prepared accordingly the method described in the “Preparation of sample solution” section. The results of all data related to precision were documented by way of relative standard deviation.
Nearness of an agreement among the value which is considered either as a conventional true value or a conventional reference value as well the value found in analytical method is called accuracy. The accuracy was estimated as the relative recovery, which was measured in three replicates at three spiked concentrations (80%, 100%, and 120%) in the sample. The placebo formulation was spiked with 0.48%, 0.60%, and 0.72% of monoammonium glycyrrhizinate and 4%, 5%, and 6% of sennoside-B, respectively, and percent recovery calculated by comparing obtained value with spiked value.
The measurement of analytical method’s ability to stay unchanged by little but intentional variation in method parameters and give an indication of its consistency while regular usage is called robustness. Robustness of the method was studied by deliberately changing the experimental conditions like column temperature and flow rate. Flow rate was changed from 1 to 0.9 mL/min and 1.1 mL/min, and column temperature varied from 40 to 38 °C and 42 to check robustness of the method.
Optimization of chromatographic conditions
Initial trials were conducted by using various mobile phase compositions containing methanol, water, and acetonitrile in different ratio but proper separation was not achieved. Various gradient methods were also tried containing these solvents but desired separation with peak shape was not achieved. Then, phosphate buffer was tried in different ratio with methanol and acetonitrile at different pH for better resolution of phytomarkers. Finalized gradient mobile phase program is shown in Table 2.
Quantification of phytomarkers
Optimized chromatographic conditions were applied to analyze sample solutions to quantify both the phytomarkers present in the formulation along with mixture of standards (Figs. 1 and 2). Individual raw materials used in the preparation of formulation, Yashtimadhu dry extract, Swarnpatri dry extract, and Aragvadha dry extract were also quantified with respective phytomarkers (Figs. 3, 4, and 5). The results calculated based on calibration curve of individual markers are summarized in Table 3.
Validation of the developed analytical method
The specificity was demonstrated by comparing chromatograms of standard and sample solutions (Figs. 6 and 7). From this study, it is clear that both the phytomarkers were clearly resolved, with no interference from the sample matrix.
The system suitability of the developed method was confirmed by calculating various chromatographic parameters such as number of theoretical plates (N), Resolution (Rs), tailing factor (Tf), and capacity factor (k′) from the chromatogram of standard solutions. Results obtained are shown in Table 4. System suitability parameters confirmed that the given chromatographic conditions were good for the method development and validation.
The linearity of calibration curves for monoammonium glycyrrhizinate and sennoside-B was established in range of 5–25 μg/mL. Representation of results obtained is given in Table 5.
Detection limit and quantitation limit
The LOD and LOQ values were determined and results obtained are shown in Table 6.
The intra-day precisions were measured six times a day and % relative standard deviation (RSD) ranged from 0.6078% to 1.5111% and inter-day precisions were measured on six consecutive days and % RSD ranged from 0.5602 to 1.1906%. Results obtained of precision study are shown in Table 7.
The accuracy of the assay method, measured as relative recovery at three concentration levels, was 98.96–101.39%, with all RSD values ≤ 2%. Recovery study results are shown in Table 8.
Robustness of the developed method was performed by deliberately altering the experimental conditions such as flow rate from 1 to 0.9 mL/min and 1.1 mL/min, and column temperature was varied from 40 to 38 °C and 42 °C. Results obtained of robustness study are shown in Table 9.
The research involves new RP-HPLC method for the simultaneous estimation of monoammonium glycyrrhizinate and sennoside-B in polyherbal formulations and plant materials. A previous study has shown that there are several methods for quantification of monoammonium glycyrrhizinate and sennoside-B individually [12,13,14,15,16,17]. However, there is no RP-HPLC method reported for the simultaneous estimation of these phytomarkers combined in any dosage forms. The main objective of the research work is to develop a fast, reliable, and cost-effective method for such estimation which requires minimal sample preparation.
On the basis of solubility data, it was clear that all the constituents were properly soluble in water. So water was selected as extracting solvent. Reverse phase column is widely used for separation of organic compounds, and all the phytomarkers present in tablets were effectively separated on reverse phase column such as C8 and C18. Hence, reverse phase column was used for method development. For better resolution, C18 column was preferred over C8 column. During the development of the method, several combinations of mobile phase with different solvents were evaluated. Finally, phosphate buffer at pH 3 with acetonitrile in gradient mobile phase was finalized to quantify both phytomarkers in an accurate manner (Table 2). Quantification of the polyherbal formulations along with its individual plant materials was done using a developed method (Table 3). The results obtained for quantification of phytomarkers in the formulation were in equivalent with the theoretical values derived based on the amount of phytomarkers present in raw material used in the preparation of the formulation indicating the accuracy of the method for routine quality control of the formulation.
This developed method was validated by determination of parameters like accuracy, linearity, precision, limit of detection, and quantification as well as robustness according to ICH guidelines . During the specificity study, both the phytomarkers were clearly resolved, with no interference from the sample matrix indicating the developed method usability for the analysis without interference (Figs. 6 and 7). System suitability were also checked using factor such as number of theoretical plates (N), Resolution (Rs), tailing factor (Tf), and capacity factor (k′). All the factors were observed in accordance with specified limits (Table 4). The system was found to be suitable for analyzing monoammonium glycyrrhizinate and sennoside-B by the described method. In linearity study, the obtained r2 > 0.999 for all standards indicate that the developed method is linear for both phytomarkers in tested range. LOD and LOQ were also calculated to get an idea about the minimum concentration required to detect and quantify the markers in the sample solution. Precision study was conducted via intra-day and inter-day precision study. Results obtained in both the study were in accordance with ICH guidelines, i.e., % RSD < 2% (Table 7). Accuracy study was performed by spiking the phytomarkers in placebo tablet at three different concentration levels. Full recovery was obtained for the compounds of interest which indicate that the method was able to completely recover markers from excipients in the formulation matrix (Table 8). A simple sample preparation method was the main reason for lesser variation of recovery. From the result of the robustness study, the method was found to be robust for routine analysis.
These results demonstrate that the developed method is linear, precise, and accurate. Moreover, the method consumes a small amount of sample and involves very simple sample preparation method for quantification of phytomarkers.
Due to complex nature of polyherbal formulation, it is important to ensure the quality of finished product based on reliable scientific methods, which have not been reported earlier. The proposed research involved in development and validation of analytical method used in quality control of polyherbal laxative formulation. The developed and validated HPLC method will help in the quality control of polyherbal laxative formulation by means of biologically active phytomarkers. The HPLC method developed for simultaneous estimation of monoammonium glycyrrhizinate and sennoside-B is accurate, precise, reproducible, and repeatable. With the growing demand for herbal drugs and increased belief in the usage of herbal medicine, the development of a reliable standardization tool will help in maintaining the quality of such important polyherbal preparations.
Availability of data and materials
All data and materials are available upon request.
High-performance liquid chromatography
International Council for Harmonization
Limit of detection
Limit of quantification
Reverse-phase high-performance liquid chromatography
Relative standard deviation
- S :
OF K, Egharevba HO, Ahmadu PO (2012) Standardization of herbal medicines - a review. Int J Biodivers Conserv 4(3):101–112
Kumar T, Chandrashekar K, Tripathi DK (2011) Standardization of gokshuradi churna: an ayurvedic polyherbal formulation. J Chem Pharm Res 3(3):742–749
Sitapara N, Buch P, Sheth NR DA (2011) Standardization of caturjata churna - an ayurvedic polyherbal formulation. J Pharm Herbal Formul 1(4):56–62
Khare CP (2007) Indian medicinal plant. Springer Science Media, New York
Yadav NP, Dixit VK (2008) Recent approaches in herbal drug standardization. Int J Integr Biol 2:198–199
Jatav VS, Singh SK, Sharma AK (2011) Recent pharmacological trends in glycyrrhiza glabra. Int J Pharm Front Res 1:170–185
Khare P, Kishore K, Sharma DK (2017) A study on the standardization parameters of cassia angustifolia. Asian J Pharm Clin Res 10(7):329–332
Ramchander JP, Jalwal P, Middha A (2017) Recent advances on senna as a laxative: a comprehensive review. J Pharmacogn Phytochem 6(2):349–353
Anonymous (2016) The ayurvedic pharmacopoeia of india, part-I, volume-IX. Government of India, Ministry of AYUSH, New Delhi, pp 92–94
Kushawaha M, Agrawal RC (2012) Biological activity of medicinal plant cassia fistula – a review. J Sci Res Pharm 1(3):7–11
Sharma DK (2017) Enumerations on phytochemical, pharmacological and ethnobotanical properties of cassia fistula linn: yellow shower. J Phytopharma 6(5):300–306
Rosenthal I, Wolfram E, Meier B (2014) An HPLC method to determine sennoside A and sennoside B in sennae fructus and sennae folium. Pharmeur Bio Sci Notes 20(14):92–102
Ghassemi-Dehkordi N, Ghanadian M, Arabha S (2014) Development of a validated HPLC method for the determination of sennoside A and B, two major constituents of cassia obovata coll. J HerbMed Pharmacol 3(2):119–124
Latha TS, Rao AR, Veeresham C (2019) Evaluation of various marketed formulations of senna by RP-HPLC and HPTLC. J Pharmacogn Phytochem 8(1):847–852
Zhou S, Zhang X, Duan X, Mei H (2015) HPLC determination of glycyrrhizic and glycyrrhetinic acids in weiyanning granule. Biomed Res 26(2):311–314
Basar N, Talukdar AD, Nahar L, Stafford A, Kushiev H, Kan A, Sarker SD (2014) A simple semi-preparative reversed-phase HPLC/PDA method for separation and quantification of glycyrrhizin in nine samples of glycyrrhiza glabra root collected from different geographical origins. Phytochem Anal 25(5):399–404
Dutta A, Bratati DE (1998) Seasonal variation in the content of sennosides and rhein in leaves and pods of cassia fistula. Indian J Pharm Sci 60(6):388
ICH Harmonized Triplicate Guideline (2005): Validation of analytical procedures: text and methodology Q2 (R1), ICH Steering Committee, Step 4 of ICH process, Retrieved form https://database.ich.org/sites/default/files/Q2_R1__Guideline.pdf.
Authors are thankful to Vasu Research Centre, a division of Vasu Healthcare Pvt. Ltd., Vadodara, Gujarat, for providing support to carry out this research work. Authors are also thankful to Parul University, Vadodara, Gujarat, for providing essential support throughout the work.
The authentic plant extracts and polyherbal formulation was obtained as gift samples from Vasu Healthcare Pvt. Ltd., Vadodara, Gujarat.
There is no funding source for this project.
Ethics approval and consent to participate
Consent for publication
The authors declare no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Patel, M., Chauhan, S., Patel, V. et al. Analytical method development and validation for simultaneous estimation of monoammonium glycyrrhizinate and sennoside-B in polyherbal laxative tablet using RP-HPLC. Futur J Pharm Sci 7, 49 (2021). https://doi.org/10.1186/s43094-021-00200-y
- Monoammonium glycyrrhizinate
- Polyherbal tablet
- Method development and validation